Among the Microwave measurement devices, a setup of Microwave bench, which contains of Microwave devices, has a noticeable place. This entire setup, with few changes, is able to amount numerous values like guide wavelength, free space wavelength, cut-off wavelength, impedance, frequency, VSWR, Klystron characteristics, Gunn diode characteristics, power measurements, etc.
The output formed by microwaves, in determining power is usually of a little value. They differ with the situation in a transmission line. There should be an equipment to measure the Microwave power, which in general will be a Microwave bench setup.
This setup is a mixture of dissimilar parts which can be observed in detail. The resulting figure openly clarifies the setup.
As the name suggests, it makes a microwave signal, in the order of a few mill watts. This uses speed modulation technique to transfer endless wave beam into mill watt power.
A Gunn diode oscillator or a Reflex Klystron tube might be an instance for this microwave signal generator.
This is the attenuator which chooses the preferred frequency and limits the output around 0 to 50db. This is variable and can be attuned according to the requirement.
This attenuator sets the quantity of attenuation. It can be agreed as a fine adjustment of values, where the readings are tested beside the values of Precision Attenuator.
This eliminates the signal that is not essential to influence the detector mount. Isolator permits the signal to pass through the waveguide only in one direction.
This is the device which procedures the frequency of the signal. With this frequency meter, the signal can be adjusted to its resonance frequency. It also gives provision to couple the signal to waveguide.
A crystal detector probe and crystal detector mount are point out in the beyond figure, where the detector is linked through a probe to the mount. This is used to demodulate the signals.
The standing wave voltmeter offers the reading of standing wave ratio in dB. The waveguide is positioned by some gap to regulate the clock cycles of the signal. Signals conveyed by waveguide are advanced through BNC cable to VSWR or CRO to measure its characteristics.
A microwave bench set up in real-time application would look as follows −
Currently, let us have a look at the significant part of this microwave bench, the slotted line.
In a microwave transmission line or waveguide, the electromagnetic field is measured as the sum of incident wave from the producer and the reflected wave to the generator. The replications show a bad fit or incoherence. The magnitude and phase of the reflected wave be influenced upon the amplitude and phase of the reflecting impedance.
The standing waves got are measured to know the transmission line inadequacies which are essential to have knowledge on impedance mismatch for effective transmission. This slotted line helps in measuring the standing wave ratio of a microwave device.
The slotted line contains of a slotted section of a transmission line, where the measurement has to be go through. It has a travelling probe carriage, to let the probe get linked everywhere required, and the ability for attaching and detecting the instrument.
In a waveguide, a slot is made at the center of the broad side, axially. A movable probe linked to a crystal detector is introduced into the slot of the waveguide.
The output of the crystal detector is related to the square of the input voltage applied. The flexible probe permits convenient and accurate measurement at its position. However, as the probe is moved along, its output is proportional to the standing wave pattern, which is formed inside the waveguide. A variable attenuator is employed here to obtain accurate results.
The output VSWR can be obtained by
The resulting figure shows the different parts of a slotted line labelled.
The parts labelled in the above figure indicate the following.
In order to get a low frequency modified signal on an oscilloscope, a slotted line with a tunable detector is employed. A slotted line carriage with a tunable detector can be used to measure the following.
A detector mount which is used to detect the low occurrence square wave modulated microwave signals the tunable detector. The resulting figure gives an idea of a tunable detector mount.
The resulting image signifies the practical application of this device. It is dismissed at the end and has an opening at the other end just as the above one.
To deliver a match among the Microwave transmission system and the detector mount, a tunable stub is often used. There are three different types of tunable stubs.
Likewise, there are fixed stubs like −
The detector mount is the final stage on a Microwave bench which is terminated at the end.
Microwave Engineering Related Interview Questions
|Digital Electronics Interview Questions||CCNA Interview Questions|
|Electronic Communications Interview Questions||MCSE Interview Questions|
|Site Management Interview Questions||Network Administrator Interview Questions|
|Microwave Engineering Interview Questions||Ethernet Interview Questions|
|RF Engineer Interview Questions||Transmission & Distribution Interview Questions|
Microwave Engineering Tutorial
All rights reserved © 2018 Wisdom IT Services India Pvt. Ltd
Wisdomjobs.com is one of the best job search sites in India.